>> What Can the MicroMapper Assistance You Do?
Fluke’s MicroMapper is a hand-held cable tester that enables network industry experts to promptly and effortlessly validate the integrity of Ethernet twisted pair cables.
In just one action, the MicroMapper can test twisted pair cabling for opens, shorts, reversed, crossed, and break up pairs.
You want to slide the switch to the “Cable” position, press the “Examination” button and the MicroMapper will then immediately scan for any present faults in your cable.
You can use the distant unit for a person-human being tests of installed cabling. It also has a tone generator functionality which can be applied with the MicroProbe cable tracer to trace cables and identify cables concealed in ceilings, partitions, floors, and bundles.
In this article is the full listing of what the MicroMapper can assist you do.
1. Exams for wiring faults and detects opens, shorts, crossed, reversed, and break up pairs. (Break up pair fault detection calls for a minimal cable size of 15.75″ (40cm))
2. Verifies shield integrity
3. MicroMapper’s Remote Identifier allows one human being to test installed cabling
4. Effortless to read fault display screen and high speed tests
5. Generation of two tones for tracing cables and locating hidden cables
6. Debug-method for in-depth fault identification effects
7. Vehicle-rest manner to decrease electric power intake
>> What Are on the MicroMapper Unit?
The Fluke MicroMapper has a very well intended user interface for quick procedure. You can use the collection of LED gentle to confirm your screening final result.
1. Pair and Defend indicator LEDs
2. Toner LED
3. “Take a look at” button
4. Remote Adapter with RJ45 jack
5. RJ45 jack for the major unit
6. Off/Cable/Toner Change
7. Fault LEDs
8. Low Battery LED
>> How to Use Fluke MicroMapper?
1. How to Use MicroMapper’s Toner Procedure
a) Slide the change on the appropriate facet of MicroMapper to the “Toner” posture
b) Link the cable to the MicroMapper’s RJ45 jack. (To send out a tone into a patch panel, join one conclude of the involved patch twine to the MicroMapper’s RJ45 jack and the other close to a jack on the panel.)
c) To deliver tone 1, press and promptly release the “Test” button.
d) To produce tone 2, push and keep the “Test” button for two seconds.
e) Use the MicroProbe to trace the linked cable. The signal reception is strongest near the cable beneath examination.
f) Slide the switch on the suitable facet to the “OFF” position to discontinue the tone. (Usually power the unit off to avert battery drain.)
2. How to Use MicroMapper Exam
a) Slide the switch on the correct facet to the “Cable” situation to convert MicroMapper on.
b) Connect one conclude of the cable to be examined on the MicroMapper’s RJ45 jack.
c) Join the other close of the cable to the MicroMapper Remote’s RJ45 jack.
d) Press the “Exam” button to perspective the outcomes.
e) The horizontal LEDs indicate the cable’s integrity standing.
Environmentally friendly: Pair or Protect is superior
Eco-friendly Flashing: Pair or Defend is faulty
No Light: Pair is open up or cable is not shielded
The vertical LEDs reveal the wiring faults and a lower battery status. Wiring faults are: Shorter, REVERSED, MISWIRE, Split PAIRS.
f) To come across out a fault on a precise pair, use MicroMapper’s diagnostic function
g) Push and hold “Check” button for extra than 2 seconds.
MicroMapper will scan each individual pair and the protect pausing and flashing each eco-friendly LED independently. If a faulty pair is detected, the corresponding fault position will blink in purple.